In the course of writing this year’s Pittcon coverage of high performance liquid
chromatography (HPLC) columns (1,2), I observed that large numbers of columns
and packing materials designed for preparative LC were introduced at the
conference. I thought this might give me the opportunity to review some
of the basics of preparative chromatography, emphasizing the role of the
column in its successful implementation.
It often has been stated (or perhaps overstated) that the column
is the heart of the liquid chromatograph. Choice of the wrong column and
mobile-phase conditions for the sample at hand can trivialize all of the
advantages of expensive, sophisticated instrumentation and data systems.
In preparative chromatography, this statement also is true. Because one
often is working with a sample that might have limited solubility in the
injection solvent or mobile phase, and injections of large volumes of sample
(for increased throughput) are used frequently, an unsuitable column–mobile-phase
combination can be even more disastrous, with possible sample precipitation
wreaking havoc on expensive wide-bore preparative columns and instrumentation.
In this article, I will explore the columns used in preparative HPLC — how
to select the appropriate mode, mobile-phase system, and operating conditions.
I will assume that the reader already has a familiarity with analytical
HPLC method development and separation optimization.
The definition of preparative chromatography always has been fuzzy
and is dependent upon the eyes of the beholder because the mass of injected
or collected sample often is dependent upon the amount available, sample
complexity, or intended use. For some, a few micrograms of material are
sufficient for further characterization or use, while for others, tens
of grams represent preparative amounts. The purity requirement and sample
throughput (that is, the amount of material or yield per unit time collected)
also must be considered.
Analytical separations often assume Langmuir-like isotherms and
that the resolution equation is obeyed strictly. In preparative HPLC, where
columns are frequently (and sometimes heavily) overloaded, the actual isotherms
and the commonly accepted relationships no longer apply quantitatively. For
example, in analytical HPLC, a commonly accepted definition of column capacity
is the mass of sample injected that causes a 10% decline in column efficiency.
In preparative chromatography, column capacity is often not so precisely
defined because the amount of sample injected could exceed this value
by an order of magnitude. Instead, overload in preparative LC is defined
as loading that no longer permits the isolation of product at the desired
purity or recovery levels (3). Column capacity also must take into account
other molecules in the sample, including the matrix, because these compete
with the analytes of interest for active sites. Remember, when we discuss
capacity, we mean the sample capacity and not the capacity factor, which
is a measure of analyte retention. The goal of a preparative purification
is the maximum production of purified product per injection.
Basics of Scaling Up from Analytical LC to Preparative LC
Scale-up from analytical LC to preparative-scale chromatography
often is time-consuming and wasteful of materials unless an optimized scale-up
strategy is employed. As depicted in Figure 1,
a recommended method-development strategy is to develop and optimize the
initial separation on an analytical size column, overload the column while
maintaining adequate separation of components of interest, then scale-up
accordingly to a preparative column of appropriate dimensions based upon
the amount of purified compound needed. Thus, the choice of analytical column
often is dictated by the availability of preparative columns containing the
same column-packing material, either in prepacked columns or in bulk media
that users can pack. Then, using the equations shown in Figure 2,
the scale-up should be linear with perhaps only minor adjustments required
to finalize the preparative method. It is highly recommended that one
ensures that both analytical and preparative columns from the same line
of packing material be readily available before beginning the preparative
method development and optimization process.
Choosing the Appropriate Mode and Stationary Phase in Preparative
In preparative scale-up, the same separation modes that were used
in analytical-scale chromatography can be employed. However, due to cost
and availability of high performance preparative packing materials, the
cost of mobile phase and mobile-phase additives, the high throughput requirements,
and the need to recover isolated fractions in a high purity state, users
often limit themselves to the more popular modes of adsorption and reversed-phase
chromatography. Size-exclusion chromatography (SEC) and ion-exchange
chromatography (IEC) sometimes are used for preparative scale-up and
purification. A typical preparative use for SEC might be for the elimination
of high molecular weight compounds or for protein purifications. The IEC
technique is used for the preparation of inorganic or organic ionic compounds
such as proteins. Both techniques are used sometimes in conjunction with
other preparative modes. Of course, chiral preparative separations can
be achieved on preparative versions and such columns can be purchased,
but the cost of such columns relegates them to the specialty column category.
Supercritical fluid chromatography has found use in large-scale purification
of enantiomeric materials.
For decades, adsorption chromatography on silica gel or other liquid–solid
media (such as alumina and kieselguhr) was the main technique for purification
of a variety of synthetic organic mixtures as well as other sample types.
However, the overwhelming popularity of reversed-phase chromatography
in the analytical world has shifted the emphasis to this mode of operation
for preparative LC. Also, many analytical HPLC users are not familiar with
the principles of adsorption chromatography; they are more comfortable
with reversed-phase chromatography and for this reason, tend to gravitate
to it when facing a preparative need.
Liquid–solid (and normal phase) HPLC uses organic solvents that
are more volatile compared with the aqueous-based solvent systems used in
reversed-phase chromatography. In addition, normal-phase solvents have a
higher solubility for many compounds, and reversed-phase chromatography often
displays a lower sample capacity for compounds due to lower solubility. For
compound recovery, the isolated fractions from a normal-phase separation
can be recovered quickly by evaporation of the volatile organic solvent.
On the other hand, at the conclusion of reversed-phase chromatography purification,
the collected solutes are in a predominantly aqueous medium. The removal
of the water to concentrate and recover the purified fraction might take
much longer or necessitate additional isolation steps such as desalting or
solid-phase extraction. The use of higher temperatures to evaporate aqueous
media can stress the isolated drug or biomolecule or even destroy it during
the concentration process. Thus, it is important to test the temperature
stability during the analytical-scale method development before scaling up
to preparative scale.
One consideration in preparative chromatography that usually is
of lower importance in analytical chromatography is the sample capacity of
the packing material. Because throughput is a primary criterion in preparative
LC, columns with higher capacity can handle more material per injection.
For adsorption chromatography, the surface area of an adsorbent dictates
the capacity. A higher surface area sorbent will allow injection of larger
sample masses than a lower surface area sorbent. In reversed-phase chromatography,
in addition to solubility of the analyte, the bonded-phase coverage determines
the sample capacity. Although one might think that the capacity of the
reversed-phase chromatography media is dictated by the alkyl chain length
of the bonded phase (for example, C18 versus C4),
it is more influenced by the bonded-phase coverage.
Surface coverage often is expressed as micromoles per meter squared.
For a typical silica-gel packing, there are roughly 8 µmol/m2
of surface silanols available for bonding. Because in adsorption chromatography
the silanol group is responsible for analyte retention, the larger the
surface area is, the more silanols that are present and the greater the
retention. In reversed-phase chromatography, the mechanism is hydrophobic
interaction between the alkyl and aryl groups on the analyte with the bonded
phase. For a typical monomeric C18 bonded phase, for steric reasons, the
bonded-phase coverage usually is in the range of 2.5–3 µmol/m2.
Due to its smaller footprint on the surface, a monomeric C8 phase might have
a slightly larger coverage and a C4 an even larger coverage still. So on
some reversed-phase packings, the coverage of a shorter chain alkyl phase
actually can exceed that of a C18 bonded phase. Thus, the amount of available
carbon for the hydrophobic interaction with the analyte actually might provide
a better measure of surface coverage than the alkyl chain length. Most manufacturers
provide the level of carbon coverage for their particular reversed-phase
packings. In addition to bonded-phase density, the density of the particles
in the packed column also can affect the sample capacity.
Resolution is seen as the most important factor in analytical chromatography
and is equally important in preparative chromatography. However, because
columns frequently are overloaded in preparative LC and peaks are broadened,
the selectivity of the separation often is a very important factor in
successfully using preparative LC. If the selectivity between two sample
components that are to be isolated is high, one can overload the column
to a much greater extent than if the selectivity is low. Thus, choice of
stationary phase can be critical to providing the best selectivity for the
components of interest. Table I
provides some rough guidelines on sample capacity for a typical reversed-phase
column as a function of a (selectivity factor). The actual sample capacity
for your individual sample components might have to be determined by trial-and-error
Particle Size and Column Dimensions
Particle size is an important parameter for analytical HPLC. Generally,
smaller particle sizes allow greater efficiency and permit the use of
shorter columns to increase separation speed. In preparative chromatography,
the particle size is important, but the column often is used in an overloaded
state, the smaller and more expensive particles of 1.8-, 3.0-, and 3.5-µm
average diameters that are used in analytical columns generally are not
used in larger-scale preparative columns. If a sample is very complex with
poor resolution (and selectivity) among compounds of interest and overloading
sometimes is difficult, then 5-µm particles frequently are used.
For well-resolved samples, larger particles of 7 and 10 µm can be
used. Sometimes, even larger particles are used, but a tradeoff of yield,
purity, and throughput must be achieved. Because pressure drop is inversely
proportional to the particle diameter squared, larger particles give lower
pressure drop, allowing higher flow rates, which in turn increases the
throughput of preparative columns. Generally, the cost of packing material
is inversely proportional to particle size, so the larger particles usually
are less expensive than the smaller particles. However, the need for high
purity for complex samples with poor resolution between the important
analytes can require small-particle packings.
Column dimensions are dictated by the amount of material per injection
that one desires to inject. The amount of sample that can be injected
increases with column internal diameter (and length), so using the equations
in Figure 2, one can calculate the column diameter that fits the sample
size required. Typically, 4.6-mm i.d. columns are used for small-scale
preparative work, 7.8-mm i.d. columns are for semipreparative work, and
21.2-mm i.d. columns are for larger scale preparative applications. Columns
with 30-mm, 50-mm, and even greater internal diameters are available for
even higher levels of scale-up. Beyond these diameters, very large-scale
preparative and process instruments capable of extremely high flow rates
(hundreds of milliliters per minute) might be required, and thus, have
a higher degree of solvent usage.
Amount of Sample Injected
The amount (mass) of sample that can be injected onto a column can
vary widely based upon its solubility in the injection solvent. A compromise
must be made between the relative strength of the injection solvent compared
with the strength of the chromatographic mobile phase. If the injection
solvent is too strong relative to the chromatography mobile phase, the
column can be volume overloaded and the sample will spread out longitudinally
on the column at the point of injection. In an isocratic mobile phase,
this band spreading might not be recoverable — bands will only get wider.
In gradient chromatography, depending upon the initial mobile-phase strength,
analytes can be refocused and still provide good efficiency. The best condition
is when the injection solvent is weaker than the chromatographic mobile
phase, because the analyte is focused at the head of the column. In some
cases, however, there can be stationary-phase overloading with the injected
sample mass exceeding the capacity of the initial packing in the column.
The concepts of volume overloading and concentration overloading are beyond
the scope of this article. See references 3 and 4 for coverage.
Table II gives a rough idea of the sample masses and volumes that might
be injected on reversed-phase columns of various dimensions. The masses
were estimated for compounds with reasonable water solubility and only serve
as guidelines. There are many experimental variables that affect the mass
that can be injected. For compounds with relatively high k values
displaying good resolution, for gradient elution separations, or for simple
sample mixtures, one might be able to inject larger amounts. If one is attempting
to separate complex samples showing poor resolution, compounds with low
solubility in the injection solvent, or if one is using a strong injection
solvent (relative to the mobile phase), then much smaller amounts could
be required to achieve adequate separation with the absence of overloading.
Choice of Mobile Phase
Often, a linear transfer of the chromatographic conditions can be
achieved during scale-up. The solvent system that will be used should
be decided during the analytical method development. Factors influencing
solvent choice are stationary-phase and mobile-phase conditions with optimum
selectivity for compounds of interest; spectroscopic characteristics of
mobile-phase solvents (that is, UV transparency, fluorescence properties,
and mass spectroscopic compatibility); volatility for easy removal from
isolated fractions; viscosity for low column back pressure; purity for
low levels of nonvolatile contaminants; good solubility properties for
maximum sample loads; and cost of solvents employed.
Not surprisingly, solvent systems in normal phase chromatography
often fulfill these criteria, but nevertheless, preparative reversed-phase
chromatography still commands the most attention. In analytical reversed-phase
chromatography, nonvolatile buffer salts frequently are used to ensure
proper pH and prevent tailing and poor peak shape. In the development
of the analytical separation destined for preparative scale-up, it is
recommended that a volatile buffer or mobile-phase additive such as ammonium
formate or ammonia be used for the initial method because removal will
be much easier in the final stages of the method. Also, if mass spectrometry
(MS) is used for confirmation, such a system will be more compatible. If
one already has an analytical method employing a nonvolatile buffer, a
volatile buffer usually can be substituted with only minor adjustments.
Expect some differences in efficiency and selectivity from this substitution.
A short list of recommended volatile buffer salts is included in Table III.
Because buffer volatility also is a primary concern with LC–MS interfacing,
buffers used for preparative chromatography often are the same. For example,
if using a phosphoric acid-based buffer at low pH, then a trifluoroacetic
acid or formic acid system can be used with preparative LC and LC–MS,
although mass spectroscopists prefer formic acid because trifluoroacetic
acid might suppress ionization of certain analytes in the MS ion source.
Examples of Scale-up from Analytical to Preparative Chromatography
Xanthines: The basic principle of scaling up from an analytical
chromatography column to a preparative column will be illustrated for
the simple separation of two xanthines: caffeine and theophylline. These
compounds easily can be separated using reversed-phase chromatography,
as depicted in Figure 3a, which shows increasing amounts (from 0.025
µg each to 500 µg each) of the two xanthines on a 150 mm X 3.0
mm Zorbax StableBond C18 analytical column. Note that the smallest injection
mass of 0.025 µg cannot be observed in Figure 3 because
full-scale absorbance is quite insensitive The separation was performed
isocratically at a flow rate of 0.6 mL/min using a water–acetonitrile mobile-phase
system without any additives. Note that as the injected sample mass increases,
there is a slight decrease in the retention time of the two compounds,
indicating that some mass overloading is occurring. Because a standard
10-mm pathlength UV flow cell was used at a wavelength of 270 nm, the
detector electronics saturated at the higher sample amounts, resulting
in the expected flattop peaks. Sometimes, one can tune the detector to
a nonoptimum spectroscopic absorption wavelength to prevent electronic
saturation and obtain better peak-shape characteristics. The two chromatographic
peaks also were well resolved, even at the highest loading, because these
two compounds displayed very good separation selectivity. Such good selectivity
predicted good overloading in the next stage of scale-up. Using the calculation
formula from Figure 2, we next determined the maximum mass that could be
loaded on our preparative column that contained the same packing material
but with column dimensions of 150 mm X 21.2 mm, the same length as our
analytical column. Figure 3b shows the preparative chromatogram for a 25-mg
injection of each xanthine onto the larger column but this time using a
shorter pathlength preparative flow cell with the UV detector. This shorter
pathlength was needed to prevent early detector electronics saturation and
allowed us to observe the eluted xanthines, even at the 25-mg injected
level. Note that the flow rate was adjusted to 25 mL/min instead of the
30 mL/min that was calculated. The separation of the two xanthines was
almost to the baseline, meaning that both compounds could be collected
at high purity.
Antibiotics: Let us consider a separation that is a bit more
complex: the scale-up of three antibiotics — dicloxacillan, cloxacillan,
and oxacillan — that also contain some low-level impurities. Their structures
are shown in Figure 4a. Using analytical columns, we first investigated
four bonded-phase options for optimum peak shape and maximum selectivity
(5) and found the best separation in the shortest time on an endcapped
Scaling up on an analytical column rather than a preparative column
saves sample, solvent, and time. By injecting increasing amounts of sample
onto the analytical column, one quickly can determine the point where
the overload begins to affect peak shape and resolution. Figure 4b
shows the injection of increasing amounts of our three antibiotics onto a
150 mm X 4.6 mm analytical column. Note that as the mass increases, there
is a point where peak overload begins to occur. At the overload point, peaks
begin to become nonsymmetrical and eventually can become very distorted,
as evidenced by the uppermost chromatogram of Figure 4b. Therefore, when
experimenting with a larger column, in some cases, one may want to scale
back somewhat on the calculated injected sample mass to maintain good peak
shape, resolution, and peak purity. On the other hand, to improve sample
load and throughput, one might want to increase the sample size and actually
overload the column until peaks start to overlap and purity is compromised.
In the case of the preparative scale-up, we used the preparative
scale-up factors equivalent to the second chromatogram from the top in
Figure 4b –– the mass that was injected just before overloading occurred
on the analytical column. The resulting scaled preparative separation is
shown in Figure 4c. The flow rate used on this preparative column
is 21.2 times faster that the flow rate used on the analytical column, and
the amount injected was also increased by the same amount. On this larger
bore column, 2.2 mg of each antibiotic could be isolated under nonoverloaded
conditions. Thus, on the preparative column, the identical separation and
analysis time of less than 9 min was achieved as with the analytical column.
Natural Products, Anthocyanins: Anthocyanins are complex
mixtures of pigments found in fruits and other plant parts. They are associated
with the colored components of red grapes, various berries, and so on. There
is a great deal of interest in anthocyanins due to their antioxidant properties
and potential health benefits. In nature, they occur most often as glycosides
and might be acetylated, which gives rise to very heterogeneous mixtures
They can be separated by reversed-phase chromatography using water (acidified
with formic acid)–methanol gradients and detected in the visible region
of the spectra due to their color.
Anthocyanins can be extracted from fruit matrices using 70% methanol
in water (6). Unfortunately, this high percentage of methanol can wreak
havoc on the chromatography if a large volume of extract is injected onto
the column. To illustrate this effect, a commercial sample of blueberries
was extracted using the standard method (6), and the extracts were injected
into a 250 mm X 21.2 mm preparative column. The effect of injection volume
can be seen clearly in Figure 5b, where volumes of extract (diluted 3:10)
as large as 5 mL were injected onto a preparative column. With the largest
injection volume, the front part of the chromatogram is affected by the
methanol in the injection solvent with peaks spreading out and resolution
lost, while the later peaks are unaffected because their capacity factors
are quite high in the initial mobile-phase composition. Nevertheless, fractions
of the first two eluted anthocyanins (identified as peaks 1 and 2 in the
chromatogram of Figure 5) were collected automatically from the preparative
column and identified by their UV and mass spectra (7) as delphinindin-3-galactoside
and delphinindin-3-glucoside, respectively. By reinjection onto an analytical
column, their purities were determined to be greater than 99% and greater
than 97%, respectively.
Successful Use of Preparative Chromatography
Many of the factors in successful analytical HPLC are prevalent
in preparative HPLC also, but some are even more of a factor. Because samples
in preparative applications often are crude mixtures, impurities can
accumulate at the head of the column and, if not removed, can cause peak
shape and retention time change. Sometimes accumulated impurities do not
affect retention but can change the column pressure, so one must watch
for increased column pressure. It is a good idea to flush the column occasionally
with increasingly stronger solvents to remove bound impurities (8). Build-up
of material in a packed column occurs most frequently when the injection
solvent is weaker than the mobile phase and is especially noticeable when
isocratic elution is used. The stronger solvent strength used in gradient
elution tends to help with removal of strongly held impurities. Silica-gel
adsorbent tends to hold onto more polar analytes, especially basic compounds,
while reversed-phase packings tend to favor more hydrophobic impurities.
Knowing the history of the preparative column is of utmost importance.
Because impurities from previous samples can show up unexpectedly when
new preparative separation conditions are developed, it is advisable to
start with a fresh column, or if this is not feasible, it is recommended
to go through a solvent washing procedure at the very least. Of course,
if the cost of the solvents required to regenerate the column is more than
that of packing or column replacement, then it makes sense to replace the
packing or replace the column. For self-packed columns, unpacking the column
and externally cleaning the loose packing material might be easier than attempting
to clean it in the packed column. Often, the first few centimeters of a
column suffer the most contamination, and removing this material and replacing
it with fresh packing can be performed relatively easily.
Since both adsorption and reversed-phase thin-layer chromatography
(TLC) plates are now available, during method development and transfer
to the preparative adsorption column, the use of TLC can be helpful for
quickly optimizing the mobile-phase composition, for noting the presence
of strongly retained compounds at the point of spotting (equivalent to
the head of the packed bed in the preparative column), and for qualitatively
monitoring the purity of collected fractions. With TLC, one can get an
idea if the collected fraction is fairly pure or if additional method development
is required. Of course, analytical HPLC can be used, but one is never certain
of the presence of strongly retained compounds that can be slowly eluted
from the column to contaminate subsequent isolated fractions.
Some special techniques for ensuring purity in preparative chromatography
that are not used in analytical liquid chromatography are heartcutting
and recycle. In heartcutting, just the middle portion of a peak of interest
is collected and those portions of the peak in which there is possible
overlap with other sample components are discarded or sent to waste. Of
course, in heartcutting, we sacrifice yield for purity. In recycle preparative
chromatography, the effective length of a column is increased by reinjecting
the column effluent that contains partially separated compounds of interest
back into the head of the column, sometimes several times. Such a technique
can be performed manually by collecting the fraction of interest and reinjecting
it into the column. In modern HPLC-based systems, recycling can be done
automatically using the alternate column-switching technique. In the latter
technique, two preparative columns connected by multiport valves are used
alternatively to effectively lengthen the time that a series of unresolved
compounds spends on the column. Details of these special techniques are
beyond the scope of this article.
Further information about preparative chromatography columns and
proper usage can be found in reference books and reviews devoted to the
principles and applications (3,4,9–12). Modern preparative instruments
coupled to high-efficiency and high-throughput columns have made the job
of purifying impure substances much easier. UV-based and mass-based fraction
collectors can selectively cut eluted peaks to offer additional fractionation
capability. The outlook for preparative chromatography looks bright, with
rugged preparative columns that can withstand many injections now the norm,
and further work in special preparative phases such as monoliths and high-capacity
sorbents will continue.
(1) R.E. Majors, LCGC 22(3), 230–242
(2) R.E. Majors, LCGC 22(4), 322–337
(3) Preparative Liquid Chromatography, vol.
38, B.A. Bidlingmeyer, Ed., (Elsevier Scientific, New York, 1987).
(4) L.R. Snyder, J.J. Kirkland, and J.L. Glajch,
Practical HPLC Method Development, 2nd ed. (Wiley & Sons, New York,
1997), pp. 50–56.
(5) W. Long and R. Majors, “Preparative HPLC Scale-Up
of Antibiotics,” Application Note, Agilent Technologies, Inc., Wilmington,
Delaware, February 2004.
(6) W. Kalt, J.E. McDonald, R.D. Ricker, and X.Lu,
Can. J. Plant Sci. 79, 617–623 (1999).
(7) Clifford Woodward, “Scale-Up of Anthocyanin
Separations and Re-Analysis of Collected Fractions on an Agilent Prep-C18
Column” Application Note, Agilent Technologies, Inc., Wilmington, Delaware,
(8) R.E. Majors, LCGC 21(1), 19–26
(9) R.P.W. Scott, Preparative Chromatography,
Chrom-Ed Book Series, [Online] available: http://www.library4science.com/.
(10) Preparative-Scale Chromatography, E.
Grushka, Ed., (Marcel Dekker, New York, 1988).
(11) “Scale-Up and Optimization in Preparative
Chromatography: Principles and Biopharmaceutical Applications,” Chromatographic
Science, vol. 88, A.S. Rathore, Ed., (Marcel Dekker, New York, 2003).
(12) Preparative and Production Scale Chromatography,
G. Ganetsos and P.E. Barker, Eds., (Marcel Dekker, New York, 1992).
Ronald E. Majors
“Column Watch” Editor Ronald E. Majors is business
development manager, Consumables and Accessories Business Unit, Agilent
Technologies, Wilmington, Delaware, and is a member of LCGC’s editorial
advisory board. Direct correspondence about this column to “Sample Prep
Perspectives,” LCGC, Woodbridge Corporate Plaza, 485 Route 1 South, Building
F, First Floor, Iselin, NJ 08830, e-mail firstname.lastname@example.org.